Monitoring asphalt quality requires good lab equipment and the results produced depend on how well you maintain that equipment. HMA Lab Supply recommends the following weekly maintenance approach:
This procedure allows you to monitor airflow through the furnace chamber. Reduced airflow means less oxygen-rich air being pulled into the chamber, which means longer burn times. With the furnace chamber COLD, turn on the furnace and hit the START button. Watch the weight display window and note numbers appearing. You should see numbers of –3 grams to –6 grams. The blower is attempting to lift the weighing platform off the four ceramic tubes that go down to the top of the scale/load cell. The higher the numbers, the cleaner your furnace blower and exhaust chamber. Lower numbers indicate a buildup of soot in the blower/exhaust system. Refer to your owner's manual for cleaning instructions.
Allow furnace to run for 30 seconds as shown in the elapsed time window and hit the STOP button. Allow furnace to continue heating for an additional three minutes then turn the furnace OFF.
Heating element check
There are four series of Thermolyne NCAT Asphalt Content Testers, 859, 945, 1087 and 1275. Series 859 and 945 have the filter assembly hanging down inside the top of the chamber. The 1087 and 1275 series have the filter assembly built in above the actual ignition chamber.
Check the first three or four digits of the serial number on the plate located on the back of your furnace to determine which element checks apply to your particular furnace.
There are four heating element plates in these models. After the furnace chamber has warmed up at the conclusion of the lift check, make sure the green power switch is turned off. Open the chamber door, and with your bare hand, touch the two side plates, the floor and the front edge of the top plate located just above the exhaust filter assembly. They should all be starting to warm up. If one or more of the plates feels cold to the touch, they should be replaced. The back wall of the chamber does not have an element plate.
There are five heating element plates in the Series 1087 furnace. You can check the four inside the chamber by touching them with your bare hand, just like the Series 859 and 945 above.
The fifth heating element is located above the chamber, and it must be checked as follows: with the chamber at regular ignition temperature (538 degrees C), turn the furnace off with the green power switch. Hold down the #7 key on the keypad and turn the furnace back on with the green power switch. In a couple of seconds the furnace will "beep" and you can remove your finger from the # 7 key.
You will now see two temperatures displayed; the chamber temp display will show the actual chamber temperature. The percent loss window will now display the actual filter element temperature. It should read 750 degrees C. If it is considerably lower, around 500 degrees C, it means that the filter heating element is burned out and it must be replaced. You can now turn the furnace back off and then back on, without holding down any other keys, and the furnace will return to the burn mode.
Additional system checks
Ignition sample cool down shortcut: The faster a sample can be cooled down after a burn, the sooner gradation can be carried out. Using a fan is pretty much out of the question, as it could blow fines out of the sample basket and affect the gradation.
An easier way is to replace the cool down plate provided with the furnace with a simple pre-cast concrete block, 12 inches by 12 inches and 2 inches-plus thick, like a patio stepping stone found at most home improvement or garden centers. It acts like a heat sink, removing heat from the baskets and the sample, usually cooling it down in about 15 minutes.